Vi-CELL XR Software: Power, flexibility and simplicity
Now compatible with Windows 10® when using version 2.06.3
Windows 10® Upgrade
To support Windows 10® OS, the Vi-CELL XR Software has been updated to version 2.06.3. Below is a summary of the changes/updates in version 2.06.3 from the previous 2.06.2 version.
Change/Update in 2.06.3
-Camera driver supports Windows 10® OS 64-bit and Windows 7® OS32-bit
-Ability to export data as .txt file type
-Removal of the ability by an Administrator to convert a user to a service user by using the user list. A service user can only be created by a service user.
-Security is ON by default the first time the Software is launched on a new installation
-Use of “invalid characters” are not accepted in text input fields
-Under the Preferences settings, editing the significant digits display in version 2.06.2 was not being applied, in version 2.06.3 the setting is saved and applied correctly
-Preferences reverted to default settings after opening and closing software has been corrected.
-Error displays when signing samples with different sample directories has been corrected.
-File not found error when files were associated with the incorrect folder has been corrected.
-Title on error messages has been corrected.
-Software allows up to 99 users and does not crash if more users are attempted.
If you are currently using an earlier software version and would like to upgrade to version 2.06.3 please contact us for BeLux.
Please note that the Vi-CELL XR Application Software 2.06 (both 2.06.1 and 2.06.2) requires MS Windows 7® operating systems 32 bit and requires AVT Guppy Camera PN A61768. All Vi-CELL XR instruments manufactured after September 2009 or later, starting with Serial Number AN36202, will have the AVT Guppy Camera. If the Vi-CELL XR instrument has a previous version camera, please contact your ANALIS local Beckman Service Engineer for information regarding a camera upgrade.